Embargoed until 4 a.m. CT/5 a.m. ET Tuesday, March 1, 2022
DALLAS, March 1, 2022 — Magnet technology is increasingly being used in portable electronic devices, such as the Apple AirPods Pro charging case, the Apple Pencil 2nd Generation and the Microsoft Surface Pen. However, if the devices are carried in pockets near the chest, and the individual has an implanted cardiac device (ICD), the magnets may interfere with the ICD’s ability to help regulate the heart, according to new research published today in Circulation: Arrhythmia and Electrophysiology, a peer-reviewed journal of the American Heart Association.
“If you carry a portable electronic device close to your chest and have a history of tachycardia (rapid heartbeat) with an ICD, strong magnets in these devices could disable your cardioverter defibrillator,” said lead study author Corentin Féry, M.Sc., a research engineer at the University of Applied Sciences and Arts Northwestern Switzerland, Institute for Medical Engineering and Medical Informatics in Muttenz, Switzerland. “Heart patients should be aware of these risks, and their doctor should tell them to be careful with these electronic devices with magnets.”
Devices and machinery with magnets exhibit a vector field or area of magnetic influence that can inhibit pulse generators for implanted ICDs and pacemakers. In ICDs, magnets can activate a switch prohibiting the ICD from delivering lifesaving shocks. Newer portable electronic devices equipped with strong magnets can disrupt the ICD operation. Earlier research on the iPhone 12 Pro Max demonstrated that its magnetic field is strong enough to interfere with the normal operation of an implanted pacemaker or ICD when held within an inch.
In this study, researchers tested the magnetic field output of the wireless charging case of the Apple AirPods Pro, the Microsoft Surface Pen and the Apple Pencil 2nd Generation. Their magnetic field strength was measured and compared to the iPhone 12 Pro Max. Using a magnetic mapper with 64 magnetic sensors, researchers measured the magnetic field strength of these products at various distances. The portable electronic devices were also placed closer and closer to five defibrillators from two representative manufacturers until a therapy deactivation occurred. These distances then constitute the minimal safety distance at which an interaction has actually taken place.
The maximum distance for a possible interaction between the portable electronic devices and the implantable cardiac devices was:
- around 2 cm (0.78 inches) away for all of the Apple products; and
- 2.9 cm (1.14 inches) away for the Microsoft Surface Pen.
While the tests results showed the maximum distance for a possible ICD interaction, researchers said for safety the minimal distance is between 0.8 cm (0.31 inches) for the iPhone 12 Pro Max and the Apple Pencil 2nd Generation, and 1.8 cm (0.71 inches) for the Microsoft Surface Pen and the opened charging case of the Apple AirPods Pro .
“The public needs to be aware of the potential risks of portable electronic devices in addition to the iPhone 12 Pro Max that may affect anyone with an ICD,” said study co-author Sven Knecht, D.Sc., a research engineer at the Cardiovascular Research Institute Basel at University Hospital Basel, University of Basel in Switzerland. “What is most concerning is that magnets are being used in more and more portable electronic devices, and with so many magnets around us, the risk to cardiac patients is even greater.”
“These devices can cause a problem when carried in your shirt or jacket pocket in front of the chest, as well as when you are lying on the couch and resting the electronic device on your chest, or if you fall asleep with the electronic device,” Féry added. “The main thing to remember is that any electronic device may be a danger, especially ones with a magnet inside.”
In the future, the researchers plan to focus on testing e-cigarettes, other pencils for tablets and other portable electronic devices for their potential magnetic interaction with cardiac devices. “With so many copycat products and accessories available, there may be a problem for the public to know which products pose increased risks,” Knecht said.
A major limitation of the study was that it was not conducted on ICDs implanted in patients. “In people with ICDs, the impact of the magnetic interaction will depend on the individual and their overall health,” Knecht said. “This research was the first step in identifying the importance of assessing some products for safety. The next step is to confirm these interactions by testing implanted devices in volunteer patients who are at the hospital for routine tests.”
The American Heart Association recommends keeping cell phones at least six inches away from ICDs or pacemakers by using it on the ear opposite from the implantation and to avoid keeping the cell phone in a front chest pocket.
“The American Heart Association and the manufacturers of pacemakers and implantable cardioverter defibrillators have long recommended that magnets be kept away from these implanted devices. A recent Journal of the American Heart Association study reported that the magnetic field induced in the receiver coil of the iPhone 12 Pro Max can result in clinically identifiable magnet interference in pacemakers and ICDs,” said N.A. Mark A. Estes, M.D., professor of medicine and director of the Clinical Cardiac Electrophysiology Fellowship Program at the Heart and Vascular Institute of the University of Pittsburgh School of Medicine, and an American Heart Association volunteer. “The current study extends observations on magnetic field interactions with even more devices containing magnets. Patients with cardiac electronic implantable devices should be instructed to keep all electronic devices that can generate a magnetic field several inches from their pacemakers or ICDs.“
Co-authors are Adrien Desombre, B.Sc.; Thomas Quirin, M.Sc.; Patrick Badertscher, M.D.; Christian Sticherling, M.D.; and Joris Pascal, D.Sc. Authors’ disclosures are listed in the manuscript.
Studies published in the American Heart Association’s scientific journals are peer-reviewed. The statements and conclusions in each manuscript are solely those of the study authors and do not necessarily reflect the Association’s policy or position. The Association makes no representation or guarantee as to their accuracy or reliability. The Association receives funding primarily from individuals; foundations and corporations (including pharmaceutical, device manufacturers and other companies) also make donations and fund specific Association programs and events. The Association has strict policies to prevent these relationships from influencing the science content. Revenues from pharmaceutical and biotech companies, device manufacturers and health insurance providers and the Association’s overall financial information are available here.
- Multimedia is available on the right column of release link https://newsroom.heart.org/news/magnets-in-newer-portable-electronic-devices-can-interfere-with-implanted-defibrillators?preview=a7d362567b8fccc91604d161061c1959
- After March 1, view the manuscript online.
- AHA news release: Magnets in iPhone® series 12 can interfere with some implanted cardiac devices (June 2021)
- AHA health information: Living With Your Implantable Cardioverter Defibrillator (ICD)
- AHA health information: Living With Your Pacemaker
- Follow AHA/ASA news on Twitter @HeartNews
- Follow news from the AHA’s Circulation: Arrythmia and Electrophysiology journal @CirculationEP
About the American Heart Association
The American Heart Association is a relentless force for a world of longer, healthier lives. We are dedicated to ensuring equitable health in all communities. Through collaboration with numerous organizations, and powered by millions of volunteers, we fund innovative research, advocate for the public’s health and share lifesaving resources. The Dallas-based organization has been a leading source of health information for nearly a century. Connect with us on heart.org, Facebook, Twitter or by calling 1-800-AHA-USA1.
Circulation Arrhythmia and Electrophysiology
Magnetic Field Measurements of Portable Electronic Devices: The Risk Inside Pockets for Patients With Cardiovascular Implantable Devices